Placing Labels on the Glass Board
The equipment to analyse is prepared with the help of technical support. The program loaded and the labels supplied.
Then, the glass board is loaded into the placement equipment.
The machine validates the position of the glass board with the help of fiducial marks.
Finally the labels are placed on the boards as defined in the placement program.
Measurement of the Placed Labels
Once the glass board is populated with labels, it is inserted in the CmController and the x,y positioning accuracy of the labels will be measured.
The positionning angle can also be measured.
The measured position is compared with the nominal position.
These datas are the basis needed for further analysis.
Analysis of the Results
The measured values are used to evaluate the positioning accuracy of the equipment.
In addition, numerical and graphical tools are available in the analysis software CmCStat.
Beside the statement does not limit itself to a simple judgement like specifications reached or failed.
Therefore, many detailed information are displayed to help in failure root cause analysis.
For example, dependencies between offsets and nozzles or angles of placement are identified.
This knowledge gives explanation on how the accuracy of the equipment can be improved.
Use of the Results to Optimize the Accuracy
Right after the analysis of the results, it is usually possible to optimize the accuracy of an equipment using correction offsets.
In the simplest case, the advise could be for example to run a calibration using the tools provided by the manufacturer.
Frequently, it's also possible to perform offsets correction directly into the machine software.
Therefore, based on measured values, direct corrections can be made.
If mechanical causes are the reason for inaccurate placement, the measuring results can be used to identify the problem by ordering the sequence with groups.
It facilitates the failure diagnosis for the maintenance service.
Finally, after applying the corrections, the effectiveness is verified with another measurement.
According to result other measures followed from examination measurements arise from it if necessary.
Depending on the results, other measurement will be done if necessary.
Based on the reached final condition, a test report is written at the end of the evaluation.
This is written usually in our office after customer visit.
Beside the detailed description of the final results, it contains the history of the necessary steps to reach the final condition.